Sepax kits are integral to our Sepax Cell Separation System. They are supplied sterile for single-use and enable a functionally-closed environment to separate cellular product into its components.
The CS-600.1 kit is designed for processing multiple cellular products (peripheral blood stem cells, cord blood, bone marrow) in order to wash cellular debris, free hemoglobin, and cryoprotectant.
|1||A-200/B Separation Chamber||220 mL spinning chamber enabling the centrifugation and transfer of processed product|
|2||Line Pressure Monitor||0.2 µm filter that connects to the Sepax pressure sensor|
|3||Stopcock Manifold||Three automatically controlled rotational valves to direct fluid flow|
|4||Initial Line||Connects the initial bag extension line via a luer port|
|5||Washing Solution Line||Connects the kit to the washing solution|
|6||By-product Bag||1000 mL PVC by-product bag to collect waste|
|7||Final Line||Connects the kit to the final bags|
|8||Double Spike Extension Line||Extension line for connecting the initial bag via a double connection|
|9||Single Spike Extension Line||Extension line for connecting the initial bag via a single connection|
|Kit / protocol||CordWash||SmartWash||CultureWash||NeatCell||Dilution||Adipose|
Store kits in a clean and dry environment without chemical or biological contamination within the following conditions:
|Mode||In operation||Storage and transport|
|Temperature||+7 to +27°C||+4 to +40°C|
|Relative humidity||30 to 75%, non-condensing||20 to 75%, non-condensing|
Shelf life: Two years from the date of manufacture.
Sterilization: Ethylene oxide.
Material: Standard medical grade PVC, PP, and other materials.
Biocompatibility: ISO 10993-1
Fluid path: Non-pyrogenic
The CS-600.1 is CE marked and complies with the applicable requirements of the 93/42/EEC Medical Device Directive. Confirm with your local regulatory authority for compliance with other regulations.
|Product Division||Laboratory Division|